Select a sub-category
Gear Pump Selection Guide
A gear pump is a type of positive displacement pump that uses rotating gears to move fluid through the pump. The main features of a gear pump include: High volumetric efficiency: Gear pumps can move a large amount of fluid with relatively small amounts of input power. Constant flow: Gear pumps can maintain a constant flow rate regardless of changes in the system's pressure or the pump's speed. Low pulsation: Gear pumps produce low levels of pulsation, which makes them well-suited for applications where smooth flow is important. Low noise level: Gear pumps generate low noise, compared to other displacement pumps, such as peristaltic pumps.
What's Peristaltic Aseptic Filling?
Peristaltic aseptic filling is a method of filling pharmaceutical products using peristaltic pumps, which are pumps that use a series of tubes or hoses to move fluids. This method is used in the pharmaceutical industry to ensure that the products being filled are free from contamination and meet strict quality and safety standards, such as intravenous solutions, injectable medications, and other sterile fluids. It is a reliable and efficient method for filling containers with sterile fluids, ensuring that the finished product is of high quality and free of contaminants.
Approaches to reduce flow pulsation on peristaltic pumps and working principles
Peristaltic pumps inherit with flow pulsation as a natural result of the way in which the pump operates by compressing the tubing. There are several approaches that can be used to reduce flow pulsation in peristaltic pumps. The principles behind these approaches to reducing flow pulsation in peristaltic pumps involve minimizing the amount of disruption to the flow of the fluid and allowing the fluid to pass through the pump smoothly and evenly. A few commonly used methods are discussed herein.
Vacuum Pump Selection Guide, Use Tips and their Applications
A vacuum pump is a mechanical device that is used to create a vacuum by removing gas molecules from a sealed volume in order to create vacuum. There are several types of vacuum pumps available, each with its own set of features and capabilities. Some common types of vacuum pumps include: rotary vane pumps, diaphragm pumps, piston pumps, scroll umps and diffusion pumps.
Ultrasonic Cell Homogenizer Working Principle and Their Applications
Ultrasonic cell homogenizers use high frequency sound waves to disrupt and homogenize cells and tissue samples. They work by generating sound waves in the ultrasonic range (typically 20 kHz to 1 MHz) and directing them through a probe or horn that is placed in contact with the sample. The sound waves cause the cells or tissue to vibrate rapidly, which breaks them down into smaller pieces. This process is known as cavitation.
NIR spectroscopy principle and selection guidelines
NIR (near-infrared) spectroscopy is a type of spectroscopic technique that uses light in the near-infrared region of the electromagnetic spectrum (wavelengths from approximately 750 nm to 2500 nm) to identify and analyze chemical compounds. NIR spectroscopy is based on the absorption of light by molecules. When a molecule absorbs a photon of light, the energy of the photon is transferred to the molecule, causing it to vibrate or rotate. The specific vibrations or rotations that a molecule undergoes depend on the chemical bonds and functional groups present in the molecule. By measuring the absorption of light at specific wavelengths, it is possible to determine the functional groups and chemical bonds present in a sample, and thus identify the chemical compounds that are present.
The Basics of Corona Discharger Treatment and Equipment Selection Guide
Corona dischargers are a type of electrical discharge device that is used to remove contaminants and impurities from various materials, such as gases, liquids, and plastics. This is achieved by creating an electrical discharge or plasma field around the material, which ionizes the contaminants and causes them to become charged. The charged particles can then be attracted to and neutralized by a collector plate or other type of electrode.
Selection Guide of Electric Rotary Valve in Precision Fluidics
A multi-channel electric rotary valve is a type of valve that is used to control the flow of fluids in precision fluidic systems. It is typically used in applications where precise control of fluid flow is required, such as in the manufacturing of semiconductors or the analysis of biological samples. The basic principle of a rotary valve is that it uses a rotor with multiple channels or openings to control the flow of fluid through a system. The rotor is mounted on a shaft and can be rotated to align the channels with inlet and outlet ports in the valve body. When the channels are aligned with the inlet and outlet ports, fluid can flow through the valve. When the channels are not aligned with the inlet and outlet ports, the flow of fluid is blocked.
Peristaltic pump selection guide and tips, applications
A peristaltic pump is a type of mechanical pump that uses rotating rollers to compress and move a flexible tube or hose, which creates suction and propels the liquid or fluid contained in the hose. This type of pump is commonly used in a variety of industries due to its ability to handle delicate and viscous fluids, as well as its ability to operate in a variety of conditions and environments.
Glove Box Selection Guidelines
A glove box is a sealed container that is designed to allow operators to perform tasks in a controlled environment without direct contact with the materials inside. Glove boxes are commonly used in a variety of applications, including handling hazardous materials, conducting experiments in a sterile environment, and working with sensitive equipment. When selecting a glove box, there are several factors to consider. These include the size and shape of the box, the materials used in its construction, the type of gloves and other protective equipment included with the box, and the specific features and accessories that are available.